It is recommend that you change your hydrocarbon pump oil after the first 100 operating hours, ensuring minimal internal wear. For subsequent changes, check the color of the oil, or test the oil to determine if it needs to be changed. Testing oil is especially useful if your process generates acids but not particles; in such cases, the oil may be degraded even though it looks clean.
Save a small blended sample of the new oil before you fill your pump. You can occasionally compare the appearance of the oil in the pump to this sample. Change the vacuum pump oil if it has an unusual colour, is dark, contains particles, or appears dirty or turbid.
If it looks like honey, the oil does not need to be changed. If it looks like molasses, change it right away.
If your pump is exposed to acids, the appearance may be proper even though the oil is contaminated; it is recommended that you test you oil on occasion. Proper maintenance will minimize internal wear and increase the life of your vacuum pump.
Hazardous process gases can concentrate in the vacuum pump, its oil and filters. Take proper safety precautions prior to the tear-down of your unit, or opening your filter housing.
Ensure that your vacuum pump has cooled down, but is still warm, prior to removing and changing oil. The cooler the temperature of your vacuum pump, the less viscous your oil will be. If you are changing the oil on a cold unit, run the pump for a few minutes; this will ensure all particles and debris are free-flowing, allowing them to drain properly.
Many applications require frequent oil changes. Allowing a pump with contaminated oil to stand for an extended period can result in corrosion. Pumps should be run every few weeks if not in use. Take proper care if shelving the pump for extended periods of time.
Rather have a professional service your vacuum pump? Complete the online service request form or Call Intlvac at (800) 959-5517.
The requirements for modern analytical instruments and systems are changing: users in laboratories a...
In many applications, particularly those with high temperature or aggressive chemical environments, ...
Leybold is the first vacuum pump manufacturer to test the diverse application possibilities of Augme...
The smallest detectable leak rate of our PHOENIX leak detectors is 5x10-12 mbarl/s. How can these sm...
In October 2017, Intlvac was honoured with the Large Business of the Year award presented by the Hal...
This is one question we get asked a lot at Intlvac. The short answer is ‘it depends’ meaning that ...
Visit our vacuum pump blog and news page to reinforce your understanding of vacuum service and how to choose the right vacuum for you.
247 Armstrong Avenue, Unit 6, Halton Hills, Ontario L7G 4X6
(905) 873-0166 (800) 959-5517 firstname.lastname@example.org
Click here to visit Intlvac Thin Film Systems website
Subscribe to receive the latest vacuum pump promotions and discounts exclusive to subscribers.